A new, simple and precise method for measuring cyclotron proton beam energies using the activity vs. depth profile of zinc-65 in a thick target of stacked copper foils
MetadataShow full item record
© 2015 Elsevier Ltd. The proton beam energy of an isochronous 18MeV cyclotron was determined using a novel version of the stacked copper-foils technique. This simple method used stacked foils of natural copper forming 'thick' targets to produce Zn radioisotopes by the well-documented (p,x) monitor-reactions. Primary beam energy was calculated using the <sup>65</sup>Zn activity vs. depth profile in the target, with the results obtained using <sup>62</sup>Zn and <sup>63</sup>Zn (as comparators) in close agreement. Results from separate measurements using foil thicknesses of 100, 75, 50 or 25µm to form the stacks also concurred closely. Energy was determined by iterative least-squares comparison of the normalized measured activity profile in a target-stack with the equivalent calculated normalized profile, using 'energy' as the regression variable. The technique exploits the uniqueness of the shape of the activity vs. depth profile of the monitor isotope in the target stack for a specified incident energy. The energy using <sup>65</sup>Zn activity profiles and 50-µm foils alone was 18.03±0.02 [SD] MeV (95%CI=17.98-18.08), and 18.06±0.12MeV (95%CI=18.02-18.10; NS) when combining results from all isotopes and foil thicknesses. When the beam energy was re-measured using <sup>65</sup>Zn and 50-µm foils only, following a major upgrade of the ion sources and nonmagnetic beam controls the results were 18.11±0.05MeV (95%CI=18.00-18.23; NS compared with 'before'). Since measurement of only one Zn monitor isotope is required to determine the normalized activity profile this indirect yet precise technique does not require a direct beam-current measurement or a gamma-spectroscopy efficiency calibrated with standard sources, though a characteristic photopeak must be identified. It has some advantages over published methods using the ratio of cross sections of monitor reactions, including the ability to determine energies across a broader range and without need for customized beam degraders.
Showing items related by title, author, creator and subject.
Parsons, Miles James Gerard (2009)Techniques of single- and multi-beam active acoustics and the passive recording of fish vocalisations were employed to evaluate the benefits and limitations of each technique as a method for assessing and monitoring fish ...
Synchrotron radiation computed tomography versus conventional computed tomography for assessment of four types of stent grafts used for endovascular treatment of thoracic and abdominal aortic aneurysmsSun, Zhonghua; Ng, C.; Sá Dos Reis, C. (2018)Background: To determine the accuracy of synchrotron radiation computed tomography (CT) for measurement of stent wire diameters for in vitro simulation of endovascular aneurysm repair by four different types of stent ...
Diel fluctuations in solute distributions and biogeochemical cycling in a hypersaline microbial mat from Shark Bay,WAPages, Anais; Welsh, D.; Teasdate, P.; Grice, Kliti; Vacher, M.; Bennett, W.; Visscher, P. (2014)Studying modern microbial mats can provide insights into how microbial communities interact with biogeochemical cycles. High-resolution, two-dimensional distributions of porewater analytes were determined in the upper ...