Aerosol inhalation from spacers and valved holding chambers requires few tidal breaths for children
MetadataShow full item record
OBJECTIVE: The goal was to determine the number of breaths required to inhale salbutamol from different spacers/valved holding chambers (VHCs). METHODS: Breathing patterns were recorded for 2- to 7-year-old children inhaling placebo from 4 different spacers/VHCs and were simulated by a flow generator. Drug delivery with different numbers of tidal breaths and with a single maximal breath was compared. RESULTS: With tidal breathing, mean inhalation volumes were large, ranging from 384 mL to 445 mL. Mean values for drug delivery with an Aerochamber Plus (Trudell, London, Canada) were 40% (95% confidence interval [CI]: 34%-46%) and 41% (95% CI: 36%-47%) of the total dose with 2 and 9 tidal breaths, respectively. Mean drug delivery values with these breath numbers with a Funhaler (Visiomed, Perth, Australia) were 39% (95% CI: 34%-43%) and 38% (95% CI: 35%-42%), respectively. With a Volumatic (GlaxoSmithKline, Melbourne, Australia), mean drug delivery values with 2 and 9 tidal breaths were 37% (95% CI: 33%-41%) and 43% (95% CI: 40%-46%), respectively (P = .02); there was no significant difference in drug delivery with 3 versus 9 tidal breaths. With the modified soft drink bottle, drug delivery. Drug delivery was not improved with a single maximal breath with any device. CONCLUSION: For young children, tidal breaths through a spacer/VHC were much larger than expected. Two tidal breaths were adequate for small-volume VHCs and a 500-mL modified soft drink bottle, and 3 tidal breaths were adequate for the larger Volumatic VHC. Copyright © 2010 by the American Academy of Pediatrics.
Showing items related by title, author, creator and subject.
Validation of methodology for recording breathing and simulating drug delivery through spacers and valved holding chambersSchultz, A.; Le Souëf, T.; Looi, K.; Zhang, Guicheng; Le Souëf, P.; Devadason, S. (2010)Background: Output from spacers (or valved holding chambers) is sensitive to changes in breathing pattern. Different spacers have unique characteristics that may influence breathing. A method used for breathing simulation, ...
Incentive device improves spacer technique but not clinical outcome in preschool children with asthmaSchultz, A.; Sly, P.; Zhang, Guicheng; Venter, A.; Le Souãf, P.; Devadason, S. (2012)Aim: To investigate the influence of an incentive device, the Funhaler, on spacer technique and symptom control in young children with asthma and recurrent wheeze. Methods: Randomised controlled trial where 132 2-6 year ...
Lung Deposition of 99mTc-Radiolabeled Albuterol Delivered through a Pressurized Metered Dose Inhaler and Spacer with Facemask or Mouthpiece in Children with AsthmaDitcham, W.; Murdzoska, J.; Zhang, Guicheng; Roller, C.; von Hollen, D.; Nikander, K.; Devadason, G. (2014)Background: Research on the use of a pressurized metered dose inhaler (pMDI) with spacer (pMDI/spacer) in children has indicated oral inhalation via the spacer mouthpiece is more efficient than the combination of oral and ...